Alpha-A Analyzer Measurement Examples
Broadband Dielectric Spectra of PVDF
The measurements in Fig. 1-4 were done with an Alpha-ANB analyzer with ZGS active sample cell test interface and a Quatro Cryosystem system for temperature control. In order to demonstrate the overall system performance PVDF was chosen as a material with not too low losses. Fig. 1, 2 show the dielectric spectrum for frequencies from 1 mHz .. 10 MHz at several temperatures.
PVDF shows α and β relaxations with tan(δ) losses in the range 0.01 .. 0.2. At higher temperatures, DC conductivity creates low frequency increase of ε''. As shown in the 3 dimensional diagrams, the data could be measured over the entire frequency and temperature range nearly without artefacts. In order to show the data more detailed, the diagrams for capacity and tan(δ) in Fig. 3, 4 were confined to the range of interest.